Introduction：Biphasic Defibrillator Hospital First Aid Portable Biphasic External Aed Defibrillator with Monitor DF-200EBiophasic defibrillator can meet all of your clinical request from Basic Life Support (BLS) to Advanced Cardiac Life Support (ACLS). Standard...
Hospital First Aid Portable Biphasic External Aed Defibrillator with Monitor
DF-200EBiophasic defibrillator can meet all of your clinical request from Basic Life Support (BLS) to Advanced Cardiac Life Support (ACLS).
ECG, AED/Manual mode, Recorder
SPO2, NIBP module, ETCO2, Pacemaker
Widely used in hospitals, physicians, health services, emergency services, aid organisations, clinics, etc.
Display Type: 8 inch high resolution LCD display
Sweep Speed: 25mm/sec
Information: HR, Lead/Pads, On/Off Alarm, SpO2, AED Functions and Prompts, Alarm Selection and Limits, Delivered Energy.
Charge Time: Less than 7 seconds with a new fully charged battery.
Energy Display: Monitor display indicates both selected and delivered energy.
Charge Prompt Type: Voice and visual prompts.
Electrode Impedance Measurement Range: 0-250 ohms.
Patient Connection: 5-lead ECG cable, or 3-lead ECG cable, paddles.
Lead Selection: Displayed on monitor, paddles, I, II, III, AVR, AVL, AVF, V.
ECG Size: 0.25, 0.5, 1, 1.5, 2, 4 cm/mV diplay on monitor.
Heart Rate: 20-300BPM.
Heart Rate Alarm: On/Off displayed on monitor, user-selectable.
Smart Alarms: Beeper/voice prompts indicate shockable rhythm.
Measurement Range: 30~100%, ± 2% between 80%~90%, others ± 5%
Alarm Range: User set high limit and low limit.
Alarm Accuracy: ± 10% within setting values.
Alarm time Accuracy: Less than 12 sec.
Paper: 50mm thermal.
Speed: 12.5mm/sec, 25mm/sec, 50mm/sec. User-selectable 6-second delay.
Printing Method: High-resolution, thermal print head.
Print-out Modes: Manual or automatic, user-configurable.
On/Off Control: Front panel and paddle.
Automatic Function: 9-seconds recording initiated by alarm activation or defibrillator charge or defibrillator discharge.
Type: Rechargeable, Ni-MH battery, 12V.
Operating Time: For a new, fully charged battery: 60 defibrillator discharges at maximum energy, or 3 hours minimum of continuous ECG monitoring.
Additional parameters will effect operating time with different functions.
AED Function: Auto analyze and charge X3 with programmable auto energy level selection, screen prompts, and voice prompts.
Shockable Rhythms: Ventricular fibrillation with amplitude >=200UV, ventricular tachycardia with rates >=140bpm,
and QRS complex wave duration >=140ms.
Charge Control: Control on device front panel, press key on paddle.
Prompts: Voice and visual prompts.
Energy Selection: Selectable at 2, 5, 7, 10, 20, 30, 50, 70, 100, 150, 200, 300, 360 joules.
Synchronized Mode: Synchronizes defibrillator pulse to patient's R-wave. "SYNC" message displayed on monitor
ADD：Room 3410, Fulizhen Building, No. 1 kefa Road, Science Park Community, Yuehai Street, Nanshan District, Shenzhen, 518057, PEOPLES REPUBLIC OF CHINA
Copyright 2008-2021 www.promed-tech.com Promed Technology Co., Limited XMLMAP PROMED